A load cell is a transducer that is used to create an electrical signal whose magnitude is directly proportional to the force being measured. The various types of load cells include hydraulic load cells, pneumatic load cells and strain gauge load cells.
This is a standard load cell for measuring weight up to 120 Kg. The most common use of this sensor is in weighing machine. Every weighing machine which shows weight has a load cell as sensing element.
The weight conversion is indirect and happens in two stages. Through a mechanical arrangement, the force being sensed deforms a strain gauge. The strain gauge measures the deformation (strain) as an electrical signal because the strain changes the effective electrical resistance of the wire. A load cell usually consists of four strain gauges in a Wheatstone bridge configuration. Load cells of one strain gauge (quarter bridge) or two strain gauges (half bridge) are also available.
The electrical signal output is typically in the order of a few millivolts and requires amplification by an instrumentation amplifier before it can be used. The output of the transducer is plugged into an algorithm to calculate the force applied to the transducer.
Load cells are used in several types of measuring instruments such as weighing scales, universal testing machines.
Method of connecting wire: I/P – Red(+),Black(-)
O/P – Green(+),White(-)
|Type||Single Point Load Cell|
|Total Precision||C3 class|
|Applications||Weighing Scales, Retail, Bench & Counting Scales|
|Rated Load||120 Kg. Max|
|Rated Output||2.0mV/V+/- 5%|
|Input Resistance||405 +/- 6 Ohm|
|Output Resistance||350 +/- 3 Ohm|
|Excitation Voltage||5-12V DC|
|Nonlinearity||0.017% Full Scale|
|Operating Temperature||-20 °C to +65 °C|
|Safe Overload||150% Full Scale|
|Ultimate Overload||200% Full Scale|